| Abstract|| |
Background: Management of open tibial diaphyseal fractures with bone loss is a matter of debate. The treatment options range from external fixators, nailing, ring fixators or grafting with or without plastic reconstruction. All the procedures have their own set of complications, like acute docking problems, shortening, difficulty in soft tissue management, chronic infection, increased morbidity, multiple surgeries, longer hospital stay, mal union, nonunion and higher patient dissatisfaction. We evaluated the outcome of the limb reconstruction system (LRS) in the treatment of open fractures of tibial diaphysis with bone loss as a definative mode of treatment to achieve union, as well as limb lengthening, simultaneously.
Materials and Methods: Thirty open fractures of tibial diaphysis with bone loss of at least 4 cm or more with a mean age 32.5 years were treated by using the LRS after debridement. Distraction osteogenesis at rate of 1 mm/day was done away from the fracture site to maintain the limb length. On the approximation of fracture ends, the dynamized LRS was left for further 15-20 weeks and patient was mobilized with weight bearing to achieve union. Functional assessment was done by Association for the Study and Application of the Methods of Illizarov (ASAMI) criteria.
Results: Mean followup period was 15 months. The mean bone loss was 5.5 cm (range 4-9 cm). The mean duration of bone transport was 13 weeks (range 8-30 weeks) with a mean time for LRS in place was 44 weeks (range 24-51 weeks). The mean implant index was 56.4 days/cm. Mean union time was 52 weeks (range 31-60 weeks) with mean union index of 74.5 days/cm. Bony results as per the ASAMI scoring were excellent in 76% (19/25), good in 12% (3/25) and fair in 4% (1/25) with union in all except 2 patients, which showed poor results (8%) with only 2 patients having leg length discrepancy more than 2.5 cm. Functional results were excellent in 84% (21/25), good in 8% (2/25), fair in 8% (2/25). Pin tract infection was seen in 5 cases, out of which 4 being superficial, which healed to dressings and antibiotics. One patient had a deep infection which required frame removal.
Conclusion: Limb reconstruction system proved to be an effective modality of treatment in cases of open fractures of the tibia with bone loss as definite modality of treatment for damage control as well as for achieving union and lengthening, simultaneously, with the advantage of early union with attainment of limb length, simple surgical technique, minimal invasive, high patient compliance, easy wound management, lesser hospitalization and the lower rate of complications like infection, deformity or shortening.
Keywords: Bone loss, fracture tibia, limb reconstruction system
MeSH terms:Tibial fracture, fractures, compound, reconstructive surgical procedure
|How to cite this article:|
Ajmera A, Verma A, Agrawal M, Jain S, Mukherjee A. Outcome of limb reconstruction system in open tibial diaphyseal fractures. Indian J Orthop 2015;49:429-35
|How to cite this URL:|
Ajmera A, Verma A, Agrawal M, Jain S, Mukherjee A. Outcome of limb reconstruction system in open tibial diaphyseal fractures. Indian J Orthop [serial online] 2015 [cited 2019 Jul 21];49:429-35. Available from: http://www.ijoonline.com/text.asp?2015/49/4/429/159638
| Introduction|| |
Tibia being a subcutaneous bone, open fractures of the tibia with massive soft tissue injury and bone loss, has become a problem for all trauma surgeons.  The specific method of treatment of these fractures is a matter of debate, with the treatment options ranging from external fixators, ring fixators, nailing, plating, tibial synostosis, free or vascularized bone grafting along with allografts or bone substitutes, all having their own set of complications. ,,, Ring fixators have given the best results in these fractures but are cumbersome both for the patient as well as the surgeon ,,,, and are technically demanding procedures. The limb reconstruction system (LRS) consists of an assembly of clamps (usually two or three) which can slide on a rigid rail and can be connected by compression-distraction units to achieve bone transport. In this prospective study, we evaluated the outcome of the LRS in the treatment of open fractures of tibial diaphysis with bone loss as a definite mode of treatment to achieve union, as well as limb lengthening, simultaneously.
| Materials And Methods|| |
Thirty prospective cases of open fractures of tibial diaphysis with bone loss treated by using the LRS to achieve union and lengthening simultaneously between 2009 and 2012 constituted the study.
The inclusion criteria were patients between 20 and 55 years, Gustilo Anderson type III A and III B with injury severity score  of at least 5 and bone loss of 4 cm or more. Closed fracture, infected gap nonunion, osteoporotic fracture, followup less than one year or metaphyseal fractures were excluded from the study. Clearance from the institutional ethical committee was taken.
Initially, primary survey done to diagnose associated injuries and prevent life threatening complications. After advanced trauma life support resuscitation, thorough irrigation of the wound and primary splintage with crammer wire done. Third generation intravenous cephalosporin was given. Preoperative workup involved taking X-rays of the entire affected leg with knee and ankle - anteroposterior (AP) and lateral views.
All patients were taken for immediate surgical debridement with removal of all dead necrotic tissues, removing the free loose bone pieces, followed by stabilization of the fracture with application of LRS system, under spinal anesthesia under a tourniquet and image intensifier in the supine position. Tourniquet was inflated only after the debridement was completed and before osteotomy.
Schanz pins of LRS were inserted by longitudinal stab incision and separating the soft tissue down to the bone by blunt dissection on the anteriomedial surface of tibia hence preventing the risk of neurological, vascular or tendon injuries. First the proximal most, Schanz screw was inserted by sequentially predrilling with an appropriate-size bit, followed by manual insertion of the Schanz screw by the T handle, which lowers the risk of thermal necrosis and pin loosening. This proximal pin was placed at least 15 mm from the joint to avoid penetration of the joint capsule and avoid the pes tendons and patellar tendon. Then the distal most, Schanz screw inserted in the same manner. The rod and assembly connected over these proximal and distal pins maintaining the leg length and this definitive fixator was used as a guide to pass the rest of the pins, to ensure that the rail is parallel to the long axis of the bone and all the remaining screws will be on the bone, in the same plane, perpendicular to the long axis of the tibia and parallel to the knee and ankle joints and aligning the tibial tuberosity with the second metatarsal. The foot and ankle were manipulated to ensure the absence of musculotendinous tethering. We used three Schanz screws each in the proximal and distal end clamps respectively and two screws in the middle mobile clamp. All Schanz screw were inserted under C arm control so that not more than about 2 mm was protruding beyond the distal cortex, as trying to back them out can cause pin loosening because of the tapering design of pins. After putting all the pins and the fixator on the limb, the pin holding nuts (clamps) were tightened. The compression-distraction unit was put on the proximal end clamp and middle clamp in the holes provided. Low energy osteotomy was done between the proximal and middle clamps using either a gigli wire or with a thin osteotomy connecting the predrilled holes. Through two incisions anteriorly and posteromedially periosteum was elevated, and a gigli saw was passed over a tape gauze subperiosteally from posteriomedial to anterior incision. With the gigli saw the bone was cut till it reached the medial cortex and then by periosteum elevator passed subperiosteally the osteotomy completed, protecting the periosteum. In none of the case, acute docking attempted. After thorough lavage and sterile dressing, the wound was left open without trying the wound coverage methods.
Postoperatively systemic antibiotic were continued for 5 days and dressing done regularly. In patients, whose bone was exposed, as the wound condition improved the patients were planned for flap rotation. In the rest of the patients, the wound was left as such to heal, with secondary intention or if required, split skin grafting was done. Distraction of the osteotomy was started after a week of the osteotomy at the rate of one fourth turn 4 times a day resulting in a total bone transport of 1 mm/day or 7 mm/week. In some patients rate of distraction was altered during the distraction phase depending on the patient compliance and the type of regenerate. Patients were encouraged to attain knee and ankle range of motion (ROM) after the application of LRS, depending on the patient pain tolerance.
This bone transport was continued till the fracture end approximated. And when the bone ends approximated, the distraction was stopped and dynamization of the LRS system done, which was followed by weight bearing mobilization by the patient as per his tolerance [Figure 1] and [Figure 2]. The dynamized LRS system was left in situ for further 15-20 weeks for consolidation of the regenerate and fracture union to occur. After the union and consolidation phase that is, till the three cortex union seen on the AP and lateral views, the LRS system was removed and the patellar tendon bearing cast was applied allowing full weight bearing, for further 6-10 weeks. Since the LRS remained for a period of about a total of 24-50 weeks, by this time in all cases the regenerate was well consolidated. Bony and functional assessment was done by Association for the Study and Application of the Methods of Illizarov (ASAMI) criteria,  based on union, infection, limb length discrepancy and deformity at the docking site and patient activity, limping, stiffness, reflex sympathetic dystrophy and pain, respectively.
|Figure 1: Immediate postoperative anterior-posterior (AP) (a) and lateral (b) view of a 35-year-old male presenting with open grade III B fracture tibia right side with 5 cm bone loss after debridement, application of limb reconstruction system (LRS) and osteotomy with three pins in proximal and distal fragments respectively and two pins in middle mobile fragment. 10 weeks followup AP (c) and lateral (d) of the patient after distraction showing approximation of the bone ends and regenerate. 28 weeks postoperative AP (e) and lateral (f) view after consolidation phase of 18 weeks with dynamized LRS showing three cortex union. 35 weeks postoperative AP (g) and lateral (h) view and clinical photographs (i and j) of a limb after patella tendon bearing cast removal showing good alignment of the limb with healing of the wound and knee range of motion|
Click here to view
|Figure 2: Followup X-rays anterior-posterior (AP) (a) and lateral (b) view of a 40 years old male with open tibial fracture with 6 cm bone loss, 12 weeks of distraction after application of limb reconstruction system (LRS) showing approximation of the bone ends and regenerate. 30 weeks postoperative AP (c) and lateral (d) view after consolidation phase of 18 weeks with dynamized LRS. 35 weeks postoperative AP (e) and lateral (f) view after the end of consolidation phases showing adequate union, regenerate and alignment|
Click here to view
| Results|| |
The results are assessed in 25 patients. 5 patients were lost to followup and hence excluded from the study. Mean age of the patients was 32.5 years (range 20-48 years). 92% (n = 23) patients were males with road traffic accident (92%) as the major cause of injury while rest (n = 2) sustained injury due to fall from height. 92% (n = 23) of the fractures were located in the middle or lower third of the tibia. Mean followup period was 15 months (range 12-19 months). Wound healed completely itself (n = 20) along with distraction - histogensis probably because of increased vascularity at the fracture site with distraction, whereas, additional skin grafting was done in 3 and fasciocutaneous flap rotation done in 2 cases only within one week as the wound condition improved.
The mean bone loss was 5.5 cm (range 4-9 cm). The mean duration of bone transport to fill this gap that is, time from application of LRS till the bones approximated after distraction, was 13 weeks (range 8-30 weeks). The mean time for LRS in place was 44 weeks (range 24-51 weeks), this is the time from application of LRS till its removal, which includes time of 1-week waiting, distraction or bone transport time to fill bone gap of 4-9 cm and 15-20 weeks time for fracture consolidation. The mean implant index was 56.4 days/cm. After three cortex union seen on AP and lateral view, removal of the LRS done and the patella tendon bearing (PTB) cast was further given for 6-10 weeks for union to consolidate and pin tracts to heal. Thus mean time to union that is, from application of LRS till removal of PTB cast was 52 weeks (range 31-60 weeks) with mean union index of 74.5 days/cm [Figure 1], [Figure 2] and [Table 1].
Bony results as per ASAMI score were excellent in 76% (n = 19), good in 12% (n = 3), fair in 4% (n = 1) and poor in 8% (n = 2). Functional results were excellent in 84% (n = 21), good in 8% (n = 2), fair in 8% (n = 2). Poor and failure was seen in none of our patients [Table 2]. Union occurred on dynamized LRS and PTB cast in all except 2 (8%) patients who developed nonunion at docking site and had poor results and required additional bone grafting and plating to achieve union. In one patient the LRS was removed earlier due to infection whereas in other the LRS was used for an extended period, but it failed to show three cortex union on LRS and hence operated for grafting and plating. At final followup leg length discrepancy of more than 2.5 cm was found in only 2 (8%) cases, rest 23 (92%) having leg length discrepancy of <2.5 cm and were able to walk without shoe raise. None of the patient had deformity more than 7° The knee ROM, at final followup was more than 120° in 84% (n = 21) cases and ankle ROM of more than 30° in 92% (n = 23) case. The most common complication encountered was pin tract infection in 5 cases, out of which 4 being superficial which healed to dressings and antibiotics. 1 patient had a deep infection which required frame removal and subsequent debridement and finally on healing of infection, the nonunion, required plating and bone grafting [Figure 3]. Aseptic pin loosening occurred in 2 cases and was managed by further advancing the tapering pins. 1 patient developed skin reaction after fixator application and had to be put on steroids and antiallergic medication for 5 days. There were no cases of neurovascular compromise, joint subluxation or refracture.
|Figure 3: Postoperative anteroposterior (AP) (a) of a patient with open grade III B fracture tibia right side with 7 cm bone loss, after 3 months application of limb reconstruction system (LRS) showing infection and nonunion. The patient underwent removal of LRS and subsequent debridement and after healing of the wound successfully treated with plate and bone-grafting AP (b) view|
Click here to view
| Discussion|| |
Open fractures of the tibia with bone loss have been a problem every trauma surgeon has to encounter. , The conventional methods of treatment like nailing or fixators with acute docking had a high rate of complications including bone loss leading to shortening, soft tissue healing problems, increased morbidity, multiple surgeries and longer hospitalization, finally increased chances of mal union and nonunions. ,,,,, The infection rate is significantly higher in patients treated with secondary nailing after primary external fixator or after delayed primary nailing. ,, Ring fixators like Illizarov have been the most reliable solution, but application of these fixators has a steep learning curve with a high rate of patient dissatisfaction. ,,,,,, LRS is designed primarily for limb lengthening and in nonunions and deformity correction, which includes different types of clamps (usually two or three) which can slide on a rigid rail and are connected with compression and distraction units. LRS uses osseous callus distraction for bone lengthening in a variety of procedures such as bone transport, simultaneous compression and distraction at different sites, monofocal lengthening, bifocal lengthening and correction of deformities with shortening.
Although number of studies are available on the use of LRS system in nonunions and deformity correction, ,,, but literature on the use of LRS system in fresh open tibial fractures with bone loss used as a definitive mode of fixation for achieving union as well as lengthening simultaneously is not documented to the best of our knowledge. Thus, we evaluated the functional outcome of LRS in 25 cases of open fractures of tibial diaphysis with mean bone loss of approximately 5.5 cm (range 4-9 cm) with a mean age of 32.5 years (range 20-48 years) with an average followup of 15 months (range 12-19 months) as a definitive method of treatment for achieving union as well as for lengthening done simultaneously.
Illizarov introduced corticotomy which preserves intramedullary and endosteal circulation. , But it has been shown that preserving periosteum, rather than medullary circulation and low energy osteotomy are the two most important determinants for good quality regenerate, both of which was followed by us by doing a two incision gigli saw osteotomy. ,
At final followup, union occurred in all patients except in two (8%) cases who had poor results due to nonunion at docking site which required additional plating and bone-grafting for union.  None of the patient had deformity, and only two patients had leg length discrepancy of more than 2.5 cm. Thus, our study showed excellent to a good result in 84% cases which was comparable to the results of other series in which bone loss was treated by other methods like McKee et al.,  Robert et al.,  Sen et al.,  Mekhail et al.  and Donnan et al.,  all of which showed excellent to good results in almost 80-90% cases.
In our study the mean bone transport was 5.5 cm which was almost similar to the other series like Robert et al. (6 cm),  Sen et al. (5 cm),  Mekhail et al. (5.7 cm),  Donnan et al. (6.4 cm),  and Bumbasirevic et al. (6.9 cm).  Union with limb length in our series was achieved in all cases except in two cases, with index procedure of application of LRS alone in the mean duration of 52 weeks with union index time of 74.5 days/cm. The mean implant index in our series was 56.4 days/cm, which was also almost similar to the other series by Robert et al. (54 days/cm),  Sen et al. (1.4 months/cm),  Mekhail et al. (2.42 months/cm),  Bumbasirevic et al. (1.48 months/cm)  Wani et al. (1.5 months/cm)  and Atef and El-Tantawy (45 days/cm). 
The most common complication, in accordance with other series, ,,,,,, was pin tract infection which was seen in 5 (20%) of our patients, 4 had a superficial infection, which healed on antibiotics and isolated pin removal whereas 1 had a deep infection severe enough to require frame removal, and subsequent debridement. Although, in this patient, the infection subsided on serial debridement, but the fracture went into nonunion and required plating and bone grafting to achieve union. In other patient, who developed nonunion, the LRS was used for an extended time for union to occur, even beyond 19 weeks, but three cortex union was not seen in this patient and hence plating and bone grafting done to achieve union. 2 cases developed pin loosening which required frame readjustment.
Illizarov remains the gold standard in such type of fractures and authors do not claim the superiority of LRS over ring fixator and do not aim to replace it with LRS, but our study shows the effectiveness of the LRS also, in these fractures with bone loss to achieve union and limb length simultaneously by average orthopedic surgeon who are not trained to apply illizarov's fixator. LRS is easy to handle both for surgeon and patient, simple surgical technique with short learning curve, light weighted monolateral simple design that helps in early mobilization of the patient and easy access to soft tissue and wound management, minimal invasive causing less morbidity and shorter hospitalization thus permitting early rehabilitation and less economic burden. The LRS system due to the high degree of stability provided by the Schanz pins placed widespread across the clamp seats and due to radial press fit of tapered pins, can be used in both rigid as well as dynamic mode which provides axial compression at fracture site, promoting the union, thus it can be used as temporary as well as definitive fixation. The limb lengthening can be achieved by LRS simultaneously by gradual osseous callus distraction and the bone formed is completely normal, and muscles, nerves and blood vessels also grow in response to the controlled lengthening, as they do during growth without or with minimal pain, as the distraction is gradual. Patient can continue to walk during treatment, thus providing axial compression to the fracture, promoting the union. Lighter weight, lower cost and less interference with visualization of the bone on radiographs also are desirable attributes of the LRS. Further, as the LRS is a monolateral fixator, which provides better patient compliance and helps in the wound management procedures like regular dressings, skin grafting and flap rotation can be done with ease compared to circumferential fixators.
Overall LRS proved to be an effective modality of treatment in cases of open fractures tibia with bone loss as definite modality of treatment for damage control as well as for achieving union and lengthening, simultaneously, with the advantage of simple surgical technique, minimal invasive, high patient compliance, easy wound management and the lower rate of complications. However further multicentric studies and RCT are suggested before establishing it as an effective modality of treatment in cases of open fractures tibia with bone loss
| References|| |
Watson JT, Anders M, Moed BR. Management strategies for bone loss in tibial shaft fractures. Clin Orthop Relat Res 1995;315:138-52.
Maurer RC, Dillin L. Multistaged surgical management of posttraumatic segmental tibial bone loss. Clin Orthop Relat Res 1987;216:162-70.
Keating JF, Simpson AH, Robinson CM. The management of fractures with bone loss. J Bone Joint Surg Br 2005;87:142-50.
Cole JD, Ansel LJ, Schwartzberg R. A sequential protocol for management of severe open tibial fractures. Clin Orthop Relat Res 1995;315:84-103.
Krappinger D, Irenberger A, Zegg M, Huber B. Treatment of large posttraumatic tibial bone defects using the Ilizarov method: A subjective outcome assessment. Arch Orthop Trauma Surg 2013;133:789-95.
Pavolini B, Maritato M, Turelli L, D′Arienzo M. The Ilizarov fixator in trauma: A 10-year experience. J Orthop Sci 2000;5:108-13.
García-Cimbrelo E, Olsen B, Ruiz-Yagüe M, Fernandez-Baíllo N, Munuera-Martínez L. Ilizarov technique. Results and difficulties. Clin Orthop Relat Res 1992;283:116-23.
Agarwal S, Agarwal R, Jain UK, Chandra R. Management of soft-tissue problems in leg trauma in conjunction with application of the Ilizarov fixator assembly. Plast Reconstr Surg 2001;107:1732-8.
Paley D. Problems, obstacles, and complications of limb lengthening by the Ilizarov technique. Clin Orthop Relat Res 1990;250:81-104.
Rajasekaran S, Naresh Babu J, Dheenadhayalan J, Shetty AP, Sundararajan SR, Kumar M, et al.
A score for predicting salvage and outcome in Gustilo type-IIIA and type-IIIB open tibial fractures. J Bone Joint Surg Br 2006;88:1351-60.
Paley D, Catagni MA, Argnani F, Villa A, Benedetti GB, Cattaneo R. Ilizarov treatment of tibial nonunions with bone loss. Clin Orthop Relat Res 1989;241:146-65.
Korkala O, Antti-Poika I, Karaharju EO. External fixation in open fractures of leg. Analysis of traps and complications of the method. Rev Chir Orthop Reparatrice Appar Mot 1987;73:637-42.
Tropet Y, Garbuio P, Obert L, Jeunet L, Elias B. One-stage emergency treatment of open grade IIIB tibial shaft fractures with bone loss. Ann Plast Surg 2001;46:113-9.
Blachut PA, Meek RN, O′Brien PJ. External fixation and delayed intramedullary nailing of open fractures of the tibial shaft. A sequential protocol. J Bone Joint Surg Am 1990;72:729-35.
Reuss BL, Cole JD. Effect of delayed treatment on open tibial shaft fractures. Am J Orthop (Belle Mead NJ) 2007;36:215-20.
Maurer DJ, Merkow RL, Gustilo RB. Infection after intramedullary nailing of severe open tibial fractures initially treated with external fixation. J Bone Joint Surg Am 1989;71:835-8.
Do H, Sadove RC. The Ilizarov method (callus distraction) in the treatment of open fractures of the tibia. J Ky Med Assoc 1992;90:74-7.
Hoffman WR, Henry SL, Seligson D. A preliminary experience: The Ilizarov external fixator. J Ky Med Assoc 1990;88:229-32.
Zheng Q, Wu HB, Li H, Pan ZJ. Tibial infected nonunion treated by internal bone transport using the mono-lateral external fixation. Zhonghua Wai Ke Za Zhi 2006;44:544-6.
Raschke MJ, Mann JW, Oedekoven G, Claudi BF. Segmental transport after unreamed intramedullary nailing. Preliminary report of a "Monorail" system. Clin Orthop Relat Res 1992;282:233-40.
Hessmann M, Rommens PM, Hainson K. Callus distraction of femur and tibia. Experiences with the mono-fixateur - indications for procedural changes. Unfallchirurg 1998;101:370-6.
Noonan KJ, Leyes M, Forriol F, Cañadell J. Distraction osteogenesis of the lower extremity with use of monolateral external fixation. A study of two hundred and sixty-one femora and tibiae. J Bone Joint Surg Am 1998;80:793-806.
Paley D. The ilizarov corticotomy. Tech orthop. 1990:5:4;41-52.
Kulkarni M. Operative techniques of ilizarov method. In: Kulkarani GS, editor. Textbook of Orthopaedics and Trauma. 1 st
ed. Jaypee. 1999. pp. 1510-22.
Eralp L, Kocaoglu M, Ozkan K, Turker M. A comparison of two osteotomy techniques of tibial lengthening. Arch orthop trauma surg. 2004;124;5;298-300.
McKee MD, Yoo DJ, Zdero R, Dupere M, Wild L, Schemitsch EH, et al.
Combined single-stage osseous and soft tissue reconstruction of the tibia with the Ilizarov method and tissue transfer. J Orthop Trauma 2008;22:183-9.
Robert Rozbruch S, Weitzman AM, Tracey Watson J, Freudigman P, Katz HV, Ilizarov S. Simultaneous treatment of tibial bone and soft-tissue defects with the Ilizarov method. J Orthop Trauma 2006;20:197-205.
Sen C, Kocaoglu M, Eralp L, Gulsen M, Cinar M. Bifocal compression-distraction in the acute treatment of grade III open tibia fractures with bone and soft-tissue loss: A report of 24 cases. J Orthop Trauma 2004;18:150-7.
Mekhail AO, Abraham E, Gruber B, Gonzalez M. Bone transport in the management of posttraumatic bone defects in the lower extremity. J Trauma 2004;56:368-78.
Donnan LT, Saleh M, Rigby AS, McAndrew A. Radiographic assessment of bone formation in tibia during distraction osteogenesis. J Pediatr Orthop 2002;22:645-51.
Bumbasirevic M, Tomic S, Lesic A, Milosevic I, Atkinson HD. War-related infected tibial nonunion with bone and soft-tissue loss treated with bone transport using the Ilizarov method. Arch Orthop Trauma Surg 2010;130:739-49.
Wani N, Baba A, Kangoo K, Mir M. Role of early Ilizarov ring fixator in the definitive management of type II, IIIA and IIIB open tibial shaft fractures. Int Orthop 2011;35:915-23.
Atef A, El-Tantawy A. Management of open infected comminuted tibial fractures using Ilizarov concept. Eur J Orthop Surg Traumatol 2014;24:403-8.
A-2, Mahavir Hospital, Footi-Khothi Square, Indore, Madhya Pradesh - 452 009
Source of Support: None, Conflict of Interest: None
[Figure 1], [Figure 2], [Figure 3]
[Table 1], [Table 2]